33220A MANUAL PDF
The A function generator and all associated options will be The Keysight Technologies A func- . Operating manual, service manual, quick. Service Guide. Publication Number (order as manual set) The Agilent Technologies A is a 20 MHz synthesized function. Buy Keysight Technologies A Function Generator 20MHz GPIB, LAN, USB A. Operating manual, service manual, quick reference guide, IntuiLink.
|Country:||Moldova, Republic of|
|Published (Last):||1 September 2010|
|PDF File Size:||18.5 Mb|
|ePub File Size:||11.74 Mb|
|Price:||Free* [*Free Regsitration Required]|
You can output the burst at a rate determined by the internal rate generator or the signal level on the rear-panel Trig In connector. See chapter 5 for a complete listing of the errors. Chapter 5 Error Messages Execution Errors Data out of range; large period limits minimum pulse width For a pulse waveform, the minimum pulse width is 20 ns for periods up to 10 seconds.
AGILENT TECHNOLOGIES A USER MANUAL Pdf Download.
For example, the following command sets the output frequency to 5 kHz: This error may be the result of a device failure or extreme conditions such as lightning or strong magnetic fields. Simplified Programming Overview Chapter 4 Remote Interface Reference Simplified Programming Overview Simplified Programming Overview This section gives an overview of the basic techniques used to program the function generator over the remote interface. Then, rotate the handle manuual the desired position.
The Mznual Event enable register is cleared when: PER command and then change the function to sine wave, the specified period will be used for the 3322a0 function. This command is similar to a Serial Poll but it is processed like any other instrument command. Instrument State Storage The function generator has five storage locations in non-volatile memory to store instrument states. The commands used for sweep are listed on page The Agilent A has a fixed series output impedance of 50 ohms to the front-panel Output connector.
Page Chapter 5 Error Messages Execution Errors Settings conflict; not able to sweep noise, sweep turned off The function generator cannot generate a sweep using the noise function. You must insert a blank space to separate a parameter from a command keyword. Select from 2 mHz to 20 kHz. In AM, the amplitude of the carrier is varied by the instantaneous voltage of the modulating waveform. Page Chapter 5 Error Messages Execution Errors Settings conflict; high level changed due to low level You can set the levels to a positive or negative value, but note that the high level must always be greater than the low level.
Chapter 3 Features and Functions Output Configuration Output Frequency As shown below, the output frequency range depends on the function currently selected.
Page 65 Chapter 3 Features and Functions Output Configuration Duty Cycle Square Waves The duty cycle of a square wave represents the amount of time per cycle that the square wave is at a high level assuming that the 33220aa is not inverted. When enabled, a TTL-compatible square waveform with either a rising edge default or falling edge is output from the Trig Out connector at the beginning of the sweep.
Note that since the rear-panel Modulation In connector is dc- coupled, you can use the A to emulate 33220q voltage-controlled oscillator VCO. The default arb name is the arbitrary waveform currently active selected with FUNC: Printed hardcopy manuals are available as an extra cost option. When not used as an input, the Trig In connector can be configured as an output to enable the A to trigger other instruments at the same time as its internal trigger occurs.
The locations are numbered 0 through 4. For periods above 10 seconds, the minimum pulse width is greater. You can delete the waveform in volatile memory or any of the four user-defined waveforms in non-volatile memory. Chapter 3 Features and Functions System-Related Operations System-Related Operations This section gives information on topics such as instrument state storage, power-down recall, error 332220a, self test, and front-panel display control.
A Service Manual – A Waveform Generator from Agilent/Keysight
Index affect on amplitude 63 PM 85 load termination 35 PWM 93 offset affect on amplitude 63 tutorial description amplitude limitations 60,local language, help 28 modulation depth AM 77,local operation LAN modulation depth, percent modulation arb waveform limitations 61, logarithmic sweep Sending a text message to the display overrides the display state as set by the DISP command. To change the security code, you must first unsecure the function generator using the old security code, and then enter a new code.
For the latest specifications, Power Supply: The default is Linear. Make sure you have read the security code rules described on page before attempting to change the security code.
The Agilent A assumes that all IP addresses and other dot- notation addresses are expressed as decimal byte values, and strips all leading zeros from these byte values. Query the condition register in this group.
Provided that the Power On state is set to Default. Page Chapter 7 Tutorial Direct Digital Synthesis The A represents amplitude values by 16, discrete voltage levels or bit vertical resolution.
Agilent Technologies 33220A User Manual
The default is Hz for all functions. Page of Go. The magnitude of a sine wave is usually specified by its peak, peak-to-peak, or root-mean- square RMS value. You can select a sine, square, ramp, pulse, or arbitrary waveform noise is allowed only in the gated burst mode and dc is not allowed.
Chapter 5 Error Messages Command Errors Program mnemonic too long A command header was received which contained more than the maximum 12 characters allowed. Select an internal or external FSK source.
Select whether the function generator manul true-high or true-low logic levels on the rear-panel Trig In connector for an externally-gated burst. Features And Functions Features and Functions This chapter makes it easy to look up all the details about a particular feature of the function generator. Page Chapter 3 Features and Functions Frequency Sweep Center Frequency and Frequency Span If desired, you can set the frequency boundaries of the sweep using a center frequency and frequency span.
SCPI commands are listed where 332220a.
Page Index definitionpulse width modulation 93 rack mounting kits 30 SCPI duty cycle deviation 97, radians command terminators modulating frequency ramp waveform language overview modulating source 98, symmetry 65, parameter types modulating waveform 95 range mznualSCPI command reference This information is not directly related to waveform generation but is an important part of operating the function generator.